[trARPES] Metis secondary electron suppression
rettig at fhi-berlin.mpg.de
Mon Jun 1 12:01:04 CEST 2020
As I mentioned, I don't understand yet how this TOF blanker box works
internally yet. The pulse generator creates 2V LVTTL pulses, and the
small box somehow gernates these ~7-8V pulses out of it. You can somehow
modify the amplitude by playing with the pulse length and repetition
parameter, but as soon as you do, you get artefact readings from the DLD.
So I don't know how much flexibility there is, but Surface Concept
should know, they built this thing, and the DLD. They should be able to
tell us, but don't seem eager to help...
Am 31.05.2020 um 12:51 schrieb Ralph Ernstorfer:
> Hi Laurenz,
> this looks excellent, thanks a lot!
> Do I understand correctly that the amplitude of the pulsing unit is
> fixed (7-8 V)? So ideally we get a modified box where the amplitude is
> adjustable. Should we ask Surface Concepts for this?
> On 30.05.2020 20:36, Laurenz Rettig wrote:
>> Dear all,
>> find attached the results of the tests regarding secondary electron
>> suppression with the pulse generator and new lens settings I did last week.
>> Best, Laurenz
>> trARPES mailing list
>> trARPES at fhi-lists.de
> Ralph Ernstorfer ernstorfer at fhi-berlin.mpg.de
> Research Group Structural and Electronic Surface Dynamics
> ERC project FLATLAND
> Fritz Haber Institute
> Faradayweg 4-6 Tel: +49 30 8413-5117
> 14195 Berlin Fax: +49 30 8413-5387
> trARPES mailing list
> trARPES at fhi-lists.de
Dr. Laurenz Rettig
Fritz Haber Institute of the Max Planck Society
Department of Physical Chemistry
Dynamics of Correlated Materials
14195 Berlin, Germany
phone: +49-(0)30-8413 5225
email: rettig at fhi-berlin.mpg.de
-------------- next part --------------
An HTML attachment was scrubbed...
More information about the trARPES